Dual Output 1 Phase AC Voltage Transducer (FPVT)
|Reference Standard ||GB/T 13850-1998(IEC688-1992) |
|Long Term stability||≤±0.2%/year.No accumulation error|
|Temperature influence||≤ 100ppm/oC|
|Response Time|| <400ms|
|Output Ripple||<0.4%(peak-peak value)|
|Input burden||Current less 0.2VA;voltage less 0.1VA|
|Frequency|| Nominal frequency ±10%|
|Maximum output load||Current output Max10V pressure drop|
|Auxiliary power||Ratedvoltage80%-120%,rated current 90%-110%|
|Auxiliary load effect || <± 0.1% |
|Allow excessive input|| Current 3rate continuous 10rate 30 secs voltageMax 2ratecontinuous |
|Dielectric strength||Input /output /power/between the crust 2KV AC 1 minIEC866 |
|Impact test||ANSIC37.90a/1973,IEC255-4 (5kV1.2/50us PulsedVoltage)|
|Calibration amplitude||Full scale least±3%,zero least±1%|
|Magnetic field effect||0.4KA/m magnetic field strength change <0.05%|
|Operating condition||Temperature -10~55oC relative humidity≤95%RHnoncondensing|
|Storage condition|| Temperature -40~70oC relative humidity ≤95% RHnoncondensing |
Main functions and Characteristics
Transforms the measured signal to a voltage, after HCNR200 linearity photodiode transforms to DC output which has extremely good linear relations with the measured signal, uses the computer aided designed, the principle is simply, circuitry designed concisely, reliability is highly and installment is conveniently.
Operating environment should not have the electric dust and gas of corrosive metal and destruction insulation existence, the altitude should less than 2,500 meters.
The Zero and CAL have been adjusted when products left factory. Please don't adjust at will.